To determine the concurrent criterion-related validity of 2 activity monitors in comparison with the criterion method of indirect calorimetry in older adults after total knee arthroplasty (TKA).
Subjects completed 9 increasingly demanding daily activities in a research laboratory; each activity was performed for 7 minutes, for a total of 80 minutes, while the activity monitors and criterion method were used concurrently.
Subjects (N=21, 67% women) had a mean age ± SD of 68±7 years and a body mass index of 29±4.
Main Outcome Measures
Energy expenditure (in kcal/min) measured by accelerometer-based and multisensor-based monitors and by a criterion method. Validity was assessed by the paired t test, intraclass correlation coefficient (ICC), and Bland-Altman plots comparing the measurements from the activity monitors with those of the criterion method.
Measurements from the accelerometer-based monitor were significantly lower than those of the criterion method across all walking and nonwalking activities. The underestimations ranged from 40% to 100%. The accelerometer-based monitor demonstrated small to moderate agreement compared with the criterion method (ICCs from 0 to .38). Measurements from the multisensor-based monitor were significantly lower than those of the criterion method during several nonwalking activities; yet, the differences were minor (2%–19%). Measurements from the multisensor-based monitor during walking activities were not different compared with the criterion method. The multisensor-based monitor demonstrated moderate to excellent agreement with the criterion method (ICCs from .48 to .81).
The multisensor-based monitor showed better criterion-related validity than the accelerometer-based monitor and should be considered as a tool to measure physical activity in individuals after TKA.
List of abbreviations:ACT (Actigraph), EE (energy expenditure), ICC (intraclass correlation coefficient), PA (physical activity), SWA (SenseWear Armband), TKA (total knee arthroplasty), WOMAC (Western Ontario and McMaster Universities Osteoarthritis Index)
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Archives of Physical Medicine and Rehabilitation
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Physical activity and functional limitations in older adults: a systematic review related to Canada’s Physical Activity Guidelines.Int J Behav Nutr Phys Act. 2010; 7: 38
- Physical activity and risk of revision total knee arthroplasty in individuals with knee osteoarthritis: a matched case-control study.J Rheumatol. 2004; 31: 1384-1390
- How active are patients undergoing total joint arthroplasty? A systematic review.Clin Orthop Relat Res. 2010; 468: 1891-1904
- Do activity levels increase after total hip and knee arthroplasty?.Clin Orthop Relat Res. 2014; 472: 1502-1511
- Accuracy of intensity and inclinometer output of three activity monitors for identification of sedentary behavior and light-intensity activity.J Obes. 2012; 2012: 1-9
- Estimating energy expenditure using accelerometers.Eur J Appl Physiol. 2006; 98: 601-612
- A novel method for using accelerometer data to predict energy expenditure.J Appl Physiol. 2006; 100: 1324-1331
- Treadmill calibration of the Actigraph GT1M in young-to-middle-aged obese-to-severely obese subjects.J Obes. 2012; 2012: 1-8
- Validity of the SenseWear® Armband to predict energy expenditure in pregnant women.Med Sci Sports Exerc. 2012; 44: 2001-2008
- Study to determine the criterion validity of the SenseWear Armband as a measure of physical activity in people with rheumatoid arthritis.Arthritis Care Res. 2013; 65: 888-895
- Low validity of the Sensewear Pro3 activity monitor compared to indirect calorimetry during simulated free living in patients with osteoarthritis of the hip.BMC Musculoskelet Disord. 2014; 15: 43
- Validation of physical activity monitors in individuals with diabetes: energy expenditure estimation by the multisensor SenseWear Armband Pro3 and the step counter Omron HJ-720 against indirect calorimetry during walking.Diabetes Technol Ther. 2013; 15: 413-418
- Validity of six activity monitors in chronic obstructive pulmonary disease: a comparison with indirect calorimetry.PLoS One. 2012; 7: e39198
- The impact of gait disability on the calibration of accelerometer output in adults with multiple sclerosis.Int J MS Care. 2011; 13: 170-176
- Accelerometer and gyroscope based gait analysis using spectral analysis of patients with osteoarthritis of the knee.J Phys Ther Sci. 2014; 26: 997-1002
- Validation study of WOMAC: a health status instrument for measuring clinically important patient relevant outcomes to antirheumatic drug therapy in patients with osteoarthritis of the hip or knee.J Rheumatol. 1988; 15: 1833-1840
- Double blind randomized controlled trial of sodium meclofenamate (Meclomen) and diclofenac sodium (Voltaren): post validation reapplication of the WOMAC Osteoarthritis Index.J Rheumatol. 1992; 19: 153-159
- Actigraph. Actisoft analysis software 4.0 user's manual. MTI Health Services, Fort Walton Beach2009: 53
- Calibration of the Computer Science and Applications, Inc. accelerometer.Med Sci Sports Exerc. 1998; 30: 777-781
- Accelerometer test-retest reliability by data processing algorithms: results from the Twin Cities Walking Study.J Phys Act Health. 2011; 8: 668-674
- Test-retest reliability of a portable monitor to assess energy expenditure.Appl Physiol Nutr Metab. 2011; 36: 339-343
- Metabolic demand and muscle activation during different forms of bodyweight supported locomotion in men with incomplete SCI.Biomed Res Int. 2014; 2014: 632765
- Physiological responses and energy expenditure during competitive fencing.Appl Physiol Nutr Metab. 2014; 39: 324-328
- Glucose metabolism in discordant monozygotic twins for cardiorespiratory fitness.Rev Paul Pediatr. 2013; 31: 77-82
- Cardiorespiratory and metabolic changes during yoga sessions: the effects of respiratory exercises and meditation practices.Appl Psychophysiol Biofeedback. 2008; 33: 77-81
- Reliability and validity of a portable metabolic measurement system.Can J Appl Physiol. 1996; 21: 109-119
- Validation of the VO2000 calorimeter for measuring resting metabolic rate.Clin Nutr. 2006; 25: 687-692
- Meaningful change and responsiveness in common physical performance measures in older adults.J Am Geriatr Soc. 2006; 54: 743-749
- Comparison of activity monitors to estimate energy cost of treadmill exercise.Med Sci Sports Exerc. 2004; 36: 1244-1251
- Evaluation of the SenseWear Pro Armband to assess energy expenditure during exercise.Med Sci Sports Exerc. 2004; 36: 897-904
- Measuring energy expenditure in cardiac patients using the Body Media Armband versus indirect calorimetry. A validation study.J Sports Med Phys Fitness. 2004; 44: 262-271
- Validity of a multi-sensor armband in estimating rest and exercise energy expenditure.Med Sci Sports Exerc. 2004; 36: 1063-1069
- Assessing energy expenditure in cancer patients: a pilot validation of a new wearable device.JPEN J Parenter Enteral Nutr. 2007; 31: 502-507
- Energy expenditure during daily activities as measured by two motion sensors in patients with COPD.Respir Med. 2011; 105: 922-929
- Evaluation of the SenseWear activity monitor during exercise in cystic fibrosis and in health.Respir Med. 2009; 103: 1511-1517
- Evaluation of a multisensor armband in estimating energy expenditure in obese individuals.Obesity (Silver Spring). 2006; 14: 2217-2223
- Achievement of steady state optimizes results when performing indirect calorimetry.JPEN J Parenter Enteral Nutr. 2003; 27: 16-20
- Validation of the ActiGraph two-regression model for predicting energy expenditure.Med Sci Sports Exerc. 2010; 42: 1785-1792
- A comparative evaluation of three accelerometry-based physical activity monitors.Med Sci Sports Exerc. 2000; 32: S489-S497
- Validation of the Kenz Lifecorder EX and ActiGraph GT1M accelerometers for walking and running in adults.Appl Physiol Nutr Metab. 2008; 33: 1155-1164
- Validity of accelerometry for measurement of activity in people with brain injury.Med Sci Sports Exerc. 2005; 37: 1474-1480
- Validity of physical activity monitors in adults participating in free-living activities.Br J Sports Med. 2010; 44: 657-664
- Measuring free-living energy expenditure and physical activity with triaxial accelerometry.Obes Res. 2005; 13: 1363-1369
- Evaluation of low-intensity physical activity by triaxial accelerometry.Obesity (Silver Spring). 2007; 15: 3031-3038
Published online: October 30, 2014
Supported by the Central Research Development Fund and by the Pepper Center Scholars Pilot Program (grant no. P30-AG024827), both from the University of Pittsburgh, Pittsburgh, PA.
© 2015 American Congress of Rehabilitation Medicine. Published by Elsevier Inc. All rights reserved.